"Name": "Storage Conditions (Product)"
"Value": "Approximately 40 to 50 minutes for standard PCR protocols
"Name": "Typical run time"
"Value": "The LightCycler® 96 Instrument is a fast and precise 96-well real-time PCR device. It provides various methods for gene expression and genetic variation analysis in mono- or multicolor (up to 4 colors*) mode. The software includes analysis modules for:
High Resolution Melting
Primary applications include gene expression analysis, gene detection, gene scanning, mutation detection, methylation analysis, miRNA research, and relative quantification of target genes. The LightCycler® 96 Instrument is designed for life science research only and is not intended for use in diagnostic procedures.
*Factory-calibrated dyes: SYBR Green I / FAM/ VIC / HEX / Yellow555; LC RED 610 / Texas Red; Cy5",
"Value": "LightCycler® 96 Sistemi, 96 örneğe kadar hızlı döngü için gerçek zamanlı bir PCR sistemidir. Uygulamalar, mutlak ve bağıl niceleme, niteliksel saptama, erime eğrisi analizi, yüksek çözünürlüklü erime ve uç nokta genotiplemeyi içerir. Hassas LightCycler® 96 Sisteminin termal homojenliği ve döngü hızı, çok kısa sürede doğru ve tekrarlanabilir sonuçlar üretir.
LightCycler® 96 Sisteminin optik algılaması, hidroliz probları ve diziden bağımsız boyalar (yani, SYBR Green I) gibi diziye bağlı probları esnek bir şekilde algılar. Multipleks ve çok renkli özellikleri, dört adede kadar farklı floresan boya kullanımına izin verir. Yeni yenilikçi LightCycler® 96 Uygulama ve Sistem Yazılımı, kolay içe ve dışa aktarma işlevleri, her çalıştırmadan sonra e-posta bildirimleri, çevrimiçi izleme esnekliği ve sunucu tabanlı ağ arabirimi ile basit bir analiz iş akışı oluşturur.
Minimum Sample Number A minimum of 5 samples is required for an analysis by the automated grouping algorithm. If 3 or more groups are expected the mandatory minimum requirement is 7 samples.
Maximum number of acquisition The instrument supports a maximum number of 2000 measurements for one HRM run. This number can't be exceeded in a \"normal\" HRM run, even when the highest acquisition rate of 25/°C is used. However, there might be applications out of range that include measurement in multiple channels that could exceed the maximum of 2000 measurements. In this case we advise to either reduce the temperature range or the acquisistion rate. The calculation for the total number of measurements (formula) is:
Number of measurements = Temperature range (°C) X Acquisition Rate (Measurements/°C) X Number of used Channels
This information is not yet included in the manual but will be added in future updates.
\nThe LightCycler® 96 Application Software is installed on a computer which can be connected to the LightCycler® 96 Instrument either via a standard network or directly, as well as on a computer that is not connected to a LightCycler® 96 Instrument. \n \n• Processor: Intel Core 2 duo 2.4 GHz \n• Memory: 2 GB • Hard disk: 250 GB \n• LAN: RJ45 Ethernet (100 MBit) \n• USB: USB 2.0 \n• Display resolution: 1280 * 1024 \n• Validated Operating system: Microsoft Windows XP, Microsoft Windows 7, 32bit configuration or Microsoft Windows 10 \n• Microsoft .NET Framework 4.0 installed \n• Setting for regional and language options: English (USA)",
"Name": "Additional Equipment and Reagent Required"
Importing MP24 sample lists
Although not mentioned in the current LightCycler® 96 Operators Guide, the Application Software also allows for the import of MagNA Pure 24 sample lists (.txt or .xml) in addition to MagNA Pure 96 sample lists. The import can be achived via the option Sample Editor -> Table View -> Import MP96 Sample List. So essentially MP24 sample list can be imported via the option that is also used to import MP96 sample lists.
Frequency of restarting the LightCycler 96 instrument
It is strongly recommended to restart the LightCycler 96 instrument once a day to avoid potential software problems even if the instrument is not used.
SMBv1 network protocol required for Automated Backup to Network Share feature The Automated Backup to Network Share feature from LightCycler® 96 uses a Network Protocol called Server Message Block (SMB) also called \"samba\". Several version of the SMB protocol exists and are used. The LightCycler® 96 Instrument is only compatible with the version 1 of the protocol (SMBv1). Disabling or blocking SMBv1 on the Network will affect the Automated Backup feature. So customers using the Automatic Backup feature must allow the SMBv1 protocol on their Network.
Instrument Software V1.2
Units Concerned Instrument Software V1.2 will be pre-installed on all instruments equippted with the new IPC (Internal PC)-Baytrail. All LightCycler® 96 with Serial Number 13426 and higher that are produced with the new IPC (Internal PC)-Baytrail including the new Instrument Software V1.2. Older Instruments will nnot and can't be updated to the new software.
Executive Summary The software version 1.2:
Does not include new features or bug fix
Does not change the specification of the instrument
Is not backwards compatible
Can only be used in combination of IPC-Baytrail
Is available on GRIPS (see below for details concerning the end-to-end protection)
Includes minor changes in the GUI
Has additional logs for the Operating System
Contains a few limitation
GUI changes 1. Introduction of an \"About\" screen. 2. The USB logo is blinking red when a USB device is in use and should not be disconnected. 3. Design of the startup splash screen has changed and is now composed of the following screens. 4. The start sequence shows the additional step \"Startup\".
Other non-visible changes 1. Roche Vanilla Agent for Axeda is updated to version 126.96.36.1990 2. The underlying Linux OS was updated. 3. The zip file created for the logs has a new folders structure
ic folder contains Instrument Control logs: No change in the content
im folder contains Instrument Management logs and error logs: No change in the content
os folder contains Linux OS and SW update logs: New logs for 2nd and 3rd level support
Distribution The Instrument Software version 1.2 is available on GRIPS in the folder \"Instrument Software Download\" and is also located on the USB drive delivered with the instrument. The package can only be used to reinstall the software on the instrument. Installing the Software version 1.2 on a LightCycler® 96 Instrument with IPC Small is not possible. The software is protected by an integrated end-to-end protection. The instrument will not install the package if it was altered.
Documentation The current \"LightCycler 96 System Manual 2.0 SW 1.1 and Addendum 1 - June 2016\" also called System Guide, is valid for Instrument Software version 1.2. The System Guide contains the following statement: \"Roche Diagnostics GmbH reserves the right to make any changes necessary without notice as part of ongoing product development.\" Moreover there were no changes in the specification
Limitations The instrument software V1.2 has the following known limitation:
It is not possible to access the instrument on the network by using the hostname. The IP address of the instrument must be used.
It is not possible to recalibrate the screen (re calibration of the screen is lost after reboot)
Instructions Instructions related to Service Software version 1.2 is available in LightCycler® 96 iSDoc V19.0 and higher. See also related documents:
The use of LightCycler® Application Software 1.1 has been validated for Windows 10. A corresponding \"Declaration of Software Validation\" has been released and has been uploaded under \"Downloads\"-\"Certificate\".
Workaround for Statistical Issue in Relative Quantification Dual Color Assays
For a Dual Color assay setup with 2 Reference results for every sample, an error calculation is done even if no replicates are present in the run if the same sample name is used for 2 different target genes. For this particular case, the customer can avoid this by naming the samples differently. Later, the customer can then create 2 Analysis, one for each gene This is the only way the software can be prevented from making a replicate statistic in case of a Dual Color application in case of more than one target gene per sample but identical reference genes for both targets.
Display issue for for certain rarely used dye combinations affecting the Endpoint Genotyping Application of Software Version 1.1
The results of the Endpoint fluorescence analysis are not correctly displayed, whenever the second dye name used in the analysis is earlier in alphabetical order than the first dye name. For all of this dye combinations (dye A - dye B), the results are displayed switched between the dyes, so that:
all wells called \"Homozygote for dye A\" are in reality \"Homozygote for dye B\" and vice versa
the scatterplot axis labeled as \"dye A\" is the axis for \"dye B\" instead and vice versa
the homozygote heatmap entries are switched between the dyes
The following very rarely used dye combinations are impacted by the issue:
FAM - Cy5
Hex - Cy5
Red610 - Cy5
TexasRed - Cy5
VIC - Cy5
VIC - Red610
Yellow555 - Texas Red
Yellow555 - Red610
Yellow555 - Cy5
The most commonly used dye combination (FAM - VIC) and all other dye combinations listed below areNOT impacted by the issue:
2 LightCycler® 96 Quick Guides: -System installation -Programming and running an experiment
1 LightCycler® 96 USB Drive containing: -Executable for the software -Release notes -License texts -LightCycler® 96 System Guides -Decontamination and shipment preparation -Demo data and experiment templates provided by Roche
1 Mains power cable (EU)
1 Mains power cable (US)
1 LAN cable (3 m)
2 Ventilation dust filters
1 Fuse package FUSE 5x20 T8AH 250V ULR/IEC
1 Sealing foil applicator
"Value": "LightCycler® 96 Sistemi, hızlı ve hassas 96 kuyulu gerçek zamanlı bir PCR cihazıdır. Tek veya çok renkli (4 renge kadar*) modunda gen ifadesi ve genetik varyasyon analizi için çeşitli yöntemler sağlar. Yazılım, aşağıdakiler için analiz modülleri içerir:
Yüksek Çözünürlüklü Eritme
Birincil uygulamalar arasında gen ekspresyon analizi, gen tespiti, gen taraması, mutasyon tespiti, metilasyon analizi, miRNA araştırması ve hedef genlerin nispi nicelemesi. LightCycler® 96 Sistemi yalnızca yaşam bilimi araştırmaları için tasarlanmıştır ve tanı prosedürlerinde kullanılması amaçlanmamıştır.
*Fabrikada kalibre edilmiş boyalar: SYBR Green I / FAM/ VIC / HEX / Yellow555; LC RED 610 / Teksas Kırmızısı; Cy5",
"Name": "Intended Use"