Purification (rack) run In a purification (rack) run, the instrument pipettes sample material from sample tubes, processes it, and transfers the eluates to the output consumables in the cooling station. The purification (rack) run is the fully automated combination of a sample transfer run and a purification (cartridge) run.
Purification (cartridge) run In a purification (cartridge) run, the processing cartridges are pre-filled with sample material, either in a previous sample transfer run, or by manual pipetting. The instrument processes the sample material, and transfers the eluates to the output consumables in the cooling station. In the case of a previous sample transfer run, open the corresponding order to load sample details.
Post elution run In a post elution run, the instrument pipettes eluates into 8-tube strips (either empty or containing manually prepipetted PCR master mix reagent). From each eluate, the instrument pipettes up to 2 eluate aliquots. Additionally, the instrument pipettes controls. Sample information from a previous purification run is loaded when reading the barcodes of the eluate output consumables. A post elution run uses the cooling station only and requires a dedicated post elution adapter.
Sample transfer run In a sample transfer run, the instrument pipettes sample material from sample tubes into processing cartridges. You can then use the sample-filled processing cartridges in a purification (cartridge) run.
The run types differ in input and output and use different instrument stations.(see Fig. 2).
"Value": "For safety related information refer to the MagNA Pure 24 Safety Manual.
Chemical reaction producing toxic gas Mixing bleach or DNA AWAY™ Surface Decontaminant with reagents can result in a chemical reaction that produces a highly toxic gas. -> Do not allow reagents containing guanidine thiocyanate to contact sodium hypochlorite (bleach) solution or acids. See also the reference from the FDA: Transport Media Safety Risk - Use Compatible Transport Media with SARS-CoV-2 Tests that Use Bleach - Letter to Clinical Laboratory Staff and Health Care Providers.",
"Name": "Safety Information"
"Value": "MagNA Pure 24 Sistemi, tam otomatik bir klinik numune ekstraksiyonu sistemidir. Çeşitli numune türlerinden, numune tüplerinden ve çoklu akış uygulamaları için farklı hacimdeki girdilerden nükleik asitleri çıkarır. Sistemin ana özellikleri şunlardır:
1 ila 24 numune için ölçeklenebilir saflaştırma
Birincil/ikincil tüp kullanımı
Barkodlar aracılığıyla envanter ve numune yönetimi
Ana bilgisayar aracılığıyla verimli sipariş yönetimi
"Name": "Product Description"
"Value": "1. MagNA Pure 24 Instrument 2. Built-in control unit with preinstalled MagNA Pure 24 operating software and purification protocols 3. MagNA Pure 24 Accessories 4. Digital User Assistance",
MagNA Pure 24 Instrument
A fully-automated clinical nucleic acid extraction system that brings you walkaway automation for up to 24 samples.